Ac Bridges and Measurement of Inductance and Capacitance:
Ac bridges and measurement of inductance and capacitance:
A bridge is a special class of instrument consisting of four arms, where three arms have known values and an unknown value of the remaining arm is calculated. The value of unknown arm is calculated in terms of known parameters of other three arms.
Normally, bridges are used for the measurement of passive circuit component such as Resistance, inductance, capacitance etc. Generally, Bridges are classified on the basis of input base, namely:
a.Dc bridge
b.Ac bridge
AC Bridges
AC Bridges consist of a source, balance detector, and four arms. In AC bridges, all the four arms consist of impedance. The AC bridges are formed by replacing the DC battery with an AC source and galvanometer by the detector of Wheatstone bridge.They are highly useful to find out inductance, capacitance, storage factor, dissipation factor etc. Now let us derive general expression for an AC bridge balance Figure given below shows AC bridge network,
Here Z_{1}, Z_{2}, Z_{3}, and Z_{4} are the arms of the bridge. Now at the balance condition, the potential difference between b, and d must be zero. From this, when the voltage drop from a to d equals to drop from a to b both in magnitude and phase.
Thus, we have from figure e_{1} = e_{2} .
From equation 1, 2 and 3 we have Z_{1}.Z_{4} = Z_{2}.Z_{3} and when impedance are replaced by admittance, we have Y_{1}.Y_{4} = Y_{2}.Y_{3}. Now consider the basic form of an AC bridge. Suppose we have bridge circuit as shown below,
In this circuit R_{3} and R_{4} are pure electrical resistance. Putting the value of Z_{1}, Z_{2}, Z_{3} and Z_{4} in the equation that we have derived above for AC bridge.
Now equating the real and imaginary parts we get
Following are the important conclusions that can be drawn from the above equations:
(a) We get two balanced equations that are obtained by equating real and imaginary parts this means that for an ac bridge both the relation (i.e.magnitude and phase) must be satisfied at the same time. Both the equations are said to be independent if and only if both equations contain the single variable element. This variable can inductor or resistor
(b) The above equations are independent of frequency that means we do not require an exact frequency of the source voltage and also the applied source voltage waveform need not be perfectly sinusoidal.
Maxwell’s Bridge(Maxwell’s inductorcapacitor bridge):
Accurate pure standard capacitors are more easily constructed than standard inductors. Consequently, it is desirable to be able to measure inductance in a bridge that uses a capacitance standard rather than an inductance standard. The Maxwell bridge (also known as the MaxwellWein bridge) is shown in Figure below. In this circuit, the standard capacitor C3 is connected in parallel with adjustable resistor R3. R1 is again an adjustable standard resistor. and R4 may also be made adjustable. L_{s} and R_{s} represent the inductor to be measured.
The Maxwell bridge is found to be most suitable for measuring coils with a low Qfactor (i.e., where wL_{s }is not much larger than R_{s}). The circuit diagram of Maxwell bridge is shown below
According to Maxwell’s bridge circuit, the fundamental ac bridge equivalent parameters are:
Substituting for all components in equation of ac bridge,
Equating the real components in above equation,
Equating the imaginary components in above equation,
So using above expressions we can measure unknown resistance R_{s }and unknown inductance L_{4}. The expression for quality factor is given by,
The above equation shows that for high quality factor , the value of R1 must be high(in the order of 10^{5 }to 10^{6}). The cost of such high value decade resistance is very high. So, it becomes impracticable for the measurement of the inductance of coil having high quality factor.
Advantage:
 Balance equations are independent if C3 and R3 are variables.
 Frequency does not affect the calculation of unknown variable.
 Simple expressions for unknown L_{s} and R_{s} in terms of known bridge elements.
 Maxwell‘s inductancecapacitance bridge is very useful for measurement of wide range of inductance at power and audio frequency.
Disadvantage:
1.The variable capacitor is expensive and less accurate. So fixed capacitor is sometimes used. In that case, balance adjustments are done by
 Varying R1 and R3, which is difficult.
 Putting an additional resistance in series with inductance under measurement and varying this resistance and R3.
2.Limited to measurement of low Q coil (1< Q< 10) also unsuitable for very low Q (Q<1)
Note: Maxwell‘s bridge is suitable for measurement of only medium Q coils.
HayInductance Bridge
It is the modification of Maxwell’s bridge .The Hay bridge circuit diagram shown in Figure below is similar to the Maxwell bridge , except that R3 and C3 are connected in series instead of parallel, and the unknown inductance is represented as a parallel LR circuit instead of a series circuit. The balance equations are found to be exact the same as those for the Maxwell bridge . It must be remembered, however, that the measured L_{p} and R_{p} are a parallel equivalent circuit. When the bridge in Figure below is balanced,
 A bridge is a special class of instrument consisting of four arms, where three arms have known values and an unknown value of the remaining arm is calculated.

Generally, Bridges are classified on the basis of input base, namely:
a.Dc bridge
b.Ac bridge
 The Maxwell bridge and Hay inductance bridge are used for measurement of inductance and capacitance.
 The Maxwell bridge is found to be most suitable for measuring coils with a low Q factor .

You scored /0
Any Questions on Ac Bridges and Measurement of Inductance and Capacitance: ?
Please Wait...
Discussions about this note
Forum  Time  Replies  Report 

SashaWhy is there an imaginary term in the ac bridge equation and what does it stand for? 
Mar 03, 2017 
0 Replies View Replies 

Nayanwhere is the measurement of capacitance? 
Jan 08, 2017 
0 Replies View Replies 